Salt spray test machine is used to test the corrosion resistance of various materials after their surfaces are treated by electroplating, anodizing, spraying and anti-rust oil anti-corrosion treatment. salt spray corrosion tester can be used to assess the ability of salt spray corrosion resistance of materials and their protective coatings, as well as the process quality comparison of similar protective coatings, and can also be used to assess the ability of salt spray corrosion resistance of some products.Equipment can reach in the experiment work space from room temperature to 55 ℃ between any one temperature point, and maintain a constant.
internal dimensions: 600 x 450 x 400(L x W x H)mm
extenal size: about 1075 * 1185 * 630(L* W * H)mm
air saturator: RT + 47 ℃ ~ 63 ℃
temperature uniformity: plus or minus 2 ℃
temperature fluctuation degree: plus or minus 0.5℃
spray deposition: 1 ~ 2ml / 80cm2 · h
Spray mode: continuous/cycle optional
Power supply: 220V single-phase 1.5kw 50HZ
Heating and spray system control
the salt spray corrosion test chamber adopts u-shaped alloy high-speed heating electric heat pipe, pure titanium heating pipe, acid-alkali corrosion resistance, long service life and completely independent system.It does not affect the salt spray test and control circuit.Temperature controlled output power is calculated by microcomputer to achieve high accuracy and efficiency.
use tower sprayer (height of tower tip can be adjusted);
the nozzle is made of special quartz glass, which can adjust the spray amount and spray Angle;
spray mode: continuous, gap optional;
test timing range: 0-999 hours, minutes, seconds, any setting.
Super large salt water tank can spray continuously for 72 hours without adding salt water two kinds of automatic and manual water adding systems, automatic or manual pressure bucket replenishment, laboratory water level, prevent water shortage and ultra-high temperature damage of the instrument.
Clear the salt mist in the test chamber when the machine stops to prevent corrosive gas from flowing out and losing other precision instruments in the laboratory test chamber overtemperature and water shortage protection
saturator overtemperature and water shortage protection test operation instruction end of test indication saturator level observation display external observation of salt fog deposition